Infuse in 3 easy steps
Wind large knob back so the black tab is at end of it’s track.
Load the prepared syringe.
Switch ON and go!
FREEDOM60® Key Features:
Made in the USA with highest level of quality control for defect-free product. No product maintenance needed.
Safe constant pressure
Highly accurate constant pressure pump for safe delivery. Maintains pressure of 13.5 psi.
DynEQ™ (Dynamic Equilibrium)
If pressure in sites increase, the system immediately responds by automatically slowing down.
Advanced technology All parts are designed to fit & work together within the pump. The Negator motor is the 'heart' of the FREEDOM60®.
Easy to use & learn
only 2 controls. Just Wind, load & go!
Recessed on/off switch
No inadvertent shut off.
Large wind knob
Ergonomic, easy to grasp & turn, disengages when the pump is ON, cannot be overwound.
No batteries or electricity needed. Comes with a travel pouch to promote patient independence.
Standard 60ml capacity
Easy to handle and purge of air. Can be filled from 1ml to 60ml.
Spring-loaded syringe holder
Locks syringe into the pump & maintains proper syringe orientation, prevents inadvertent removal during infusion.
Easy to check infusion progress, protects the syringe
Robust design with a molded ABS plastic housing. No sensitive electric parts
Watch this 6-minute training video of a home sub-q infusion demonstration.
Loading & Unloading the Syringe
Avoid damaging your FREEDOM60 with this quick tutorial on how to properly load and unload a 60 ml syringe.
Frequently Asked Questions
Is my FREEDOM60® working? How can I tell if I am infusing?
First, verify that your tubing is the right flow rate. Then, check that the pump winds properly and makes a distinctive 'whirr' noise when turned on – this means the pump is working properly. Finally, note the time the infusion starts and watch that the syringe plunger moves in accordance with your estimated infusion time. If there is any doubt, please talk to your healthcare provider.
How do I clean the FREEDOM60®?
You can clean the surface of the pump with warm water and detergent or use any surface disinfectant compatible with acetyl-butyl-styrene (ABS) and polycarbonate (PC) plastics, such as household bleach or hydrogen peroxide. Avoid the use of alcohol or alcohol containing compounds, as these tend to make ABS plastic brittle. Wipe again with clean water to rinse. The clear view syringe shield is not a removable safety feature, as it is an essential part of the pump, but it may be cleaned with a bottlebrush if absolutely necessary.
Can the FREEDOM60® be used by children?
Yes. The FREEDOM60® is so safe and easy to use it can even be used by children. In fact, its very first users were children with cystic fibrosis. Because the pump is lightweight, durable, and easy to use they found it ideal. They loved the portability and were happy to be out of a hospital environment.
Which syringe can I use with my Freedom60 pump?
Becton Dickinson & Co. BD® Luer-Lok® 60ml (US Reference #309653; EU Reference #300865) and Sherwood Medical Monoject® 60ml (Reference #8881-560125) are approved syringes.
I only need small syringes; why should I buy the FREEDOM60® when I only want to use 20-30ml of fluid?
The FREEDOM60® was designed to be both economical and versatile to accommodate a variety of infusion needs. It safely and effectively infuses anywhere from 1ml to 60ml in a standard 60ml syringe. If you are looking to use a pump that specifically accomodates 20 or 30ml syringes, see our FreedomEdge™
What if I want to administer a larger or smaller amount of fluid than 60ml?
If a dose is larger than 60ml, you can load an additional syringe because the tubing is reusable. For smaller volumes, the FREEDOM60® will adjust to whatever amount is contained in the syringe. The syringe can be filled anywhere from a full 60ml to as little as 1ml. You can also try our FreedomEdge™, which accomodates both 20 and 30ml syringes.
Can the FREEDOM60® be used to administer Vancomycin?
Yes. Many infusion centers are using Vancomycin up to 1.5 grams, delivering in 1:20 or 2:00 hours (45ml/hr and 30ml/hr sets, respectively) in a PICC or central line. The advantage of the FREEDOM60® is its constant pressure mechanism, so the rate will not dramatically change during an infusion. Ask for clinical studies.
Can the FREEDOM60® be used to administer Desferal?
Yes. The FREEDOM60® can deliver from 500mg to 4,000+mg in a variety of time frames.
Can the FREEDOM60® be used for multi-dose administrations?
No, not at this time. We do have a design for enabling the administration of multiple doses, but it has not yet been placed into production.
What happens if I turn the large wind-up knob during an infusion?
Nothing. The wind-up knob is disengaged while the pump is in the on position. There is no effect whatsoever by turning the knob in either direction.
Why is FREEDOM60®’s 13.5 psi pressure so significant?
World literature suggests that no IV pressure should ever exceed 15psi for safety of the patient. Some devices use pressures that exceed 40psi, a level unsafe for most PICC’s and certainly unsafe for veins. A controlled infusion with a steady pressure below 15psi is considered to be an effective, safe flow profile for SCIg and antibiotic IV infusions.
What affects the flow rate of the FREEDOM60®?
The two main factors are the diameter of the tubing and the viscosity of the fluid. Immune Globulin, for example, will flow much slower than an antibiotic. Since temperature also effects the viscosity of fluid, when the temperature is cold, the flow slows down and when the temperature is warm, the flow rate increases. Tubing length and the efficiency of the needle set also can influence flow rate. There are also some affects due to differences in syringes, but these generally tend to be relatively minor.
How do I stop the flow in the case of an emergency?
There is a slide clamp on the tubing set that will stop flow immediately. If the pump is turned off without using the slide clamp, some flow will continue until pressure from the black tab is removed by winding the knob clockwise to relieve all pressure in the syringe.
Since the flow rate is determined by the tubing set, what will happen if I put a standard bore IV set on the syringe and place it in the FREEDOM60®?
RMS Precision Flow Rate Tubing™ uses a custom luer disc connector which locks onto the syringe. If an attempt is made to use tubing without a special connector, the pump will not allow the syringe to be properly seated. This is an important safety feature which helps ensure use of flow control tubing designed for the performance profile of the FREEDOM60®. A standard bore IV set would not have the special luer disc connector. Therefore, if you tried to use a syringe hooked up to a standard bore IV set, the FREEDOM60® pump would reject the syringe and tubing. Another benefit of the luer disc is that, if the tubing is placed on a surface, it prevents connector contact and the resulting contamination.
How does the FREEDOM60® maintain such a steady flow without the use of batteries or electricity?
The FREEDOM60® is powered by a negator motor, which is a specially designed spring similar to the ones used to drive clocks and pocket-watches. The negator motor creates a constant, safe pressure on the syringe. The user winds up the negator spring to reset the pump for each infusion.
What if the pump is dropped? Submerged? Abused?
Check to make certain there is no visible damage to the pump. Turn the pump on without the syringe and see if it runs. Let it run to the end of its cycle and listen for the clicking sound. If you hear that sound, the pump is usable. If submerged, or if the pump shows clear signs of abuse, we recommend that the pump be replaced. If needed, the pump can be returned to the factory for overhaul/refurbishment.
What happens if I fall asleep using the FREEDOM60®?
Not a problem. Unlike electronic devices, the FREEDOM60® holds full pressure on the syringe even after the infusion is over. The natural venous blood pressure is less than 0.5psi and is no match for the 13.5psi rating of the pump. In other words, no blood can back flow into the line to cause the line to clot. After the patient is awake, they can flush the line normally and should have no problem with clotted lines.
Can the FREEDOM60® be used at high altitudes, such as in aircraft?
Yes. It is not sensitive to ambient pressures and makes an ideal patient transport IV pump.
What if the vein does collapse or clot on a patient using the FREEDOM60®?
The pump will deliver whatever it can at the safe pressure of 13.5psi. It may slow down significantly or completely stop, but it will never exceed the 13.5psi pressure inherent in the design, limiting any tissue damage.
How can I tell if the FREEDOM60® is working?
We recommend that the patient monitor use by taking note of the time the infusion started and verifying the rate of the tubing set being used. For example, if the patient is infusing a non-viscous medication (such as an antibiotic) and using the F120 set, then the plunger will move 10ml’s in five minutes, meaning that a full 60ml syringe will take 30 minutes to administer. If the plunger does not move accordingly, it is time to check on the infusion.
What is the noise that the pump creates when you first turn it on?
That is the pump looking for the syringe. The pump is designed to search for the syringe, which could be filled to any level, down to 1ml if you wanted. Once the pump finds the syringe plunger it makes a clicking sound and is then totally silent for the entire infusion. During normal flow, the pump makes no noise whatsoever.
The pump runs fine by itself, but when I connect it to a “Y” set it won’t flow. What’s wrong with the pump?
If the pump is working on its own, then the problem is in the "Y" set. First, check to see that all slide clamps are open. Then, check to make sure that the needle free adapters are compatible. Generally, the cause is a blocked or unconnected line, or total clot.
I have tested the pump on the bench with water and it flowed much faster than the stated 60ml/hr; what's going on?
The FREEDOM60® is designed to operate on a patient, with a PICC line, and with the viscosity of a drug present. When operating without the resistance of the PICC line or the back pressure of a vein, and with water, a 60ml set will flow at nominally 72ml/hr. See the manual for details on how to bench test the FREEDOM60®.
Didn't find what you were looking for? Contact our Support Team
Get high resolution images of the FREEDOM60®
Interested in Distributing FREEDOM60®?
If you would like to become a distributor, contact us!Where to Buy FREEDOM60® Become a Distributor